Exporting Images and Creating/Viewing Reports: Chison ECO3 Training Part 5 of 6
You can export images from the Chison ECO 3 to USB or a network/DICOM/PACS system. Reporting is straightforward and can be fairly powerful as it allows you to add images easily from the patient imaging archive.
This fifth video in the Chison ECO 3 ultrasound training series requires knowledge provided in Part 1 and Part 2 of this video training series.
Complete Chison ECO 3 Ultrasound Training Links:
Part 1: System Overview of the Chison ECO 3
Part 2: ECO 3 Imaging
Part 3: Doppler and M-Mode on the Chison ECO 3
Part 4: Measurements and Annotations
Part 5: Exporting and Reports on the ECO 3
Part 6: Chison ECO 3 Customization and Setup
Transcript to learn how to Export and create reports on the Chison ECO3 Ultrasound Machine:
Now let’s take a look at a report by hitting this little report page right here. And it will generate the report and any information that you took on that exam, any calculations will show up on there. I didn’t take any real velocity measurements or anything to sort, so that all we have right here is the volume to access management, if I’d taken some in Doppler or something of that sort, it would appear there.
And then I can click on images here to enter into the report. Obviously, the cine loop, with the little film strip, isn’t going to go on the report. The image of it will go onto the report. And you’ll see how it is entered onto there. And I can scroll through the images by clicking on left and right keys here. So from there, we can send via diacom structured reporting, export pring, or just save that report, and it’ll save it to that patient’s file. So save, complete. And click up here and exit that screen.
Finally, we’ll show how to export the exams. We’re going to click on this archive button right here. And here it shows the archive for this patient. If I want to see previous exams or send them, I will hit on that archive. But here I’ll want to select the images that I want to send. And I have a USB stuck in there, so I’m just going to go ahead and hit send images here. All this will depend on what it does, depending on how your system is set up, and the system set up, which we’ll get to in the next chapter. Go ahead and hit send image. And here I can choose diacom storage. I do not have anything set up in there. That’s set up someplace else. I do have a USB flash drive in. So I can hit U disc, choose bitmap or jpeg for those images, and I click export.
At this time, any of the loops that you export will be in a special format. And you want to talk to your sales person about getting the viewer that’ll allow you to view the cine loops on a PC. At this version, the cine loops are in the proprietary Terason format and you need their special viewer. The images, the static image, will come in jpeg or bitmap, whichever you choose, and you can view those on any PC without any particular reader.
So here I can continue the exam or go to a new exam, go to the archive, and select a previous exam, and I can also send those exams the same way by hitting send examine and it’ll send everything in that folder to the U disc or the diacom, depending on how you’re set up. And of course, you’d hit export. This will send the entire patient folder when you click on this, including the reports. So typically, if you’re sending over diacom or exporting, this is typically what you’d want to do, unless you’re just saving very specific images that you’d want to do.
You can also search for patient by name up top. So test. If I just type test, it appears right there. That’s the one I’m doing. And it’ll say, how far back do you want me to go? All patients, one year, six months, couple days, et cetera. Multiple choice here. I’m just going to go back to that full patient list. If I check multiple choice, it’ll let me select multiple patients. And then I can also speak export those to USB.
So unclick that. But going back to a patient, if we want to restart a new exam on this patient, like we talked about before, we can open up that patient, hit continue exam, which if you’ve ended the exam, this will not appear. But you can review the images, do a new exam on that patient. So if I do that here’s that patient’s information, and I would start a new study or just click OK. And there we have all that. Next, we’ll get into the system set up and connectivity.